Calibration and Repair of Test Instruments
Download the Equipment Service Form (Also fillable in your browser)
Calibration of Digital Pressure Gauges
Frequency of calibration
DG-1000’s, DG-700’s and DG-500’s should be recalibrated every two years to maintain the instruments accuracy specifications of 1% of reading, or 0.15 Pa, whichever is greater. DG-2’s and DG’3 should be recalibrated every year.
Calibration locationCalibration is performed at our facility in Minneapolis, MN (see address below). We do not calibrate gauges at conferences or trade shows.
CostCalibration of a TEC digital pressure gauge is $125. This price includes return ground shipping within the continental 48 states and Canada (expedited or air shipments will cost more). Customers in Canada – Non-warranty repairs are dutiable and GST applies to the value of the repair only. The invoice must include the original value of goods and the repair amount. Brokerage fees apply to shipments sent UPS Standard to Canada.
Other informationTurn around time at our facility for calibration is typically 2-3 days following receipt of the instrument (assuming no repairs are necessary). We will provide a NIST traceable calibration certificate which includes “as found” as well as “current condition” pressure data.
How to get your gauge calibratedPlease complete the Equipment Service Form and include it with the equipment being sent to us. (NOTE: The Equipment Service Form now contains entry fields and can be filled out digitally on your computer. If you fill it out digitally simply print the completed form. This form can still be printed and completed by hand as well. If you are trying to fill out the PDF online using FireFox, click here.)
DG-700/500 Reset ProcedureOn occasion, when a new set of batteries are installed in a DG-700 or DG-500 Pressure Gauge, the Gauge will appear as if it has locked up. This is characterized by looking at the display after it has been turned on and all of the icons and digits stay on the display and the buttons do not work. Another condition that requires a reset is when the Gauge turns on, shows the battery voltage, then shows Pr/Pr and immediately turns off. The reset process is extremely simple –
- Turn the Gauge off
- Open up the battery compartment door on the back of the Gauge
- Remove all batteries
- Press and hold the ON button for a slow count of 20
- Reinstall batteries and install battery cover
- Turn on Gauge
You may have to repeat the process one more time. If the Gauge does not return to normal operation it may need servicing that can only be performed at TEC. Complete the Equipment Service Form above and send along with the Gauge.
Field calibration checkIn lieu of sending your DG-1000, DG-700 or DG-500 gauge into TEC once every two years for calibration, customers may choose to perform a field calibration check procedure which is contained in the Digital Gauge Field Calibration Check document. This document describes a procedure to field check the calibration of digital gauges against a recently calibrated digital gauge that is being used as an in-house reference standard. We recommend that a field calibration check be performed annually, or whenever a gauge has been dropped or damage is suspected. Results of field calibration checks should be recorded and documented by the customer using the sample log sheets provided at the end of the calibration check document. Gauges that fail the Digital Gauge Field Calibration Check will need to be sent to TEC for recalibration.
Comparing channels on the same gaugeIt is also a good idea to periodically check the two channels of the DG-1000, DG-700 and DG-500 pressure gauge against each other to be sure they are reading very close to same value (e.g. they should always be within 1% of each other). This test will quickly isolate gauges with large problems (e.g. damaged flow sensors), and is described in the Comparing Channels on the Same Gauge document. This procedure does not replace the need for gauge recalibration once every two years.
Calibration of Blower Door and Duct Blaster® FansTEC Blower Door and Duct Blaster® fans maintain their calibration unless physical damage occurs to the fan or flow sensing system. Conditions which can degrade calibration of the fan are easily detected and should be field tested by the customer on a regular basis using the procedures contained in the Fan Field Check document (see Fan Field Check document). This document describes a procedure to check for damaged flow sensors, leaks in the flow sensor or tubing running from the flow sensor to the fan pressure tap, or improper positioning of the flow sensor relative to the fan housing. We recommend that a field check of the fan be performed at least annually, or whenever a fan has been dropped or damage is suspected. Results of field checks should be recorded and documented and sample log sheets are provided at the end of the field check document for this purpose.
Blower Door and Duct Blaster fans can be custom calibrated at the TEC facility for a fee. Custom fan calibration parameters can be easily entered into our TECTITE™ and TECBLAST™ software. Note: Custom fan calibration is not necessary for most testing applications.
Field Calibration Plate for Duct Blaster System with DG-1000 or DG-700 GaugeThe Field Calibration Plate allows you to quickly check the calibration of the entire Duct Blaster System with DG-1000 or DG-700 gauge. The calibration plate, which contains a 5 inch diameter hole, is simply mounted on the square transition piece that is connected to the end of the flex duct. The Duct Blaster fan is turned on to pull air through the calibration plate and simulate a duct leakage test. Results from the simulated test are compared to the known leakage rate to determine if the entire system is within a +/- 3% accuracy range. This calibration check covers the entire system, including the Duct Blaster fan, the DG-1000 or DG-700 gauge, and the flex duct.
A Duct Blaster Field Calibration Plate can be purchased from us. Instructions for using the Calibration Plate are contained in the Duct Blaster Field Calibration Plate document.
Calibration of APT Data Acquisition BoxesTEC recommends that APT Data Acquisition Boxes be recalibrated once per year at our facility in Minneapolis, MN (see address below). The cost to calibrate an APT Data Acquisition Box is $160. This price includes return ground shipping within the continental 48 states and Canada (expedited or air shipments will cost more). Turn around time at our facility for calibration is typically 3-4 days following receipt of the instrument (assuming no repairs are necessary).
When returning an APT Data Acquisition Box for calibration, please use the Equipment Service Form, fill it out completely, and include a copy of the completed form along with the equipment being returned.
Returning EquipmentClick here for instructions on how to properly return equipment to us.
Equipment RepairsIf you ever have an equipment problem, we always stock a complete line of replacement parts, and can respond quickly to any service problem. All equipment needing repairs (or calibration) should be sent to our facility at:
The Energy Conservatory
- 2801 21st Ave. South
Minneapolis, MN 55407
- (612) 827-1117 phone
(612) 827-1051 fax
Attn: Equipment Repair/Calibration
When sending equipment for calibration or repairs, please use the Equipment Service Form, fill it out completely, and include a copy of the completed form along with the equipment being returned.
For product information click on Products in the main navigation.